Report Type:	Expression Report
Date:          	10:00PM 05/17/2007
______________________________________________________________________

Filename:		Soybean EM-Axis-Parenchyma1.CHP
Probe Array Type:	Soybean
Algorithm:		Statistical
Probe Pair Thr:	8
Controls:		Antisense
______________________________________________________________________

Alpha1:		0.05
Alpha2:		0.065
Tau:		0.015
Noise (RawQ):	1.540
Scale Factor (SF):	10.412
TGT Value:		500
Norm Factor (NF):	1.000
______________________________________________________________________

Background:
	Avg: 53.48	Std: 0.81	Min: 50.80	Max: 55.10
Noise:
	Avg: 2.17	Std: 0.06	Min: 2.00	Max: 2.40
Corner+
	Avg: 372		Count: 32
Corner-
	Avg: 11473		Count: 32
Central-
	Avg: 12533		Count: 9
______________________________________________________________________

The following data represents probe sets that exceed the probe pair threshold 
and are not called "No Call".

Total Probe Sets:	61078
Number Present:	17022	27.9%
Number Absent:	43317	70.9%
Number Marginal:	739	1.2%

Average Signal (P):	3368.6
Average Signal (A):	71.0
Average Signal (M):	277.8
Average Signal (All):	992.5
______________________________________________________________________

Spike Controls:
Probe Set		Sig(5')	Det(5')	Sig(M')	Det(M')	Sig(3')	Det(3')	Sig(all)	Sig(3'/5')
AFFX-BIOB		2116.9	P	3148.3	P	2068.0	P	2444.40	0.98
AFFX-BIOC		6810.7	P	 	 	5246.7	P	6028.68	0.77
AFFX-BIOD		11573.7	P	 	 	23132.1	P	17352.91	2.00
AFFX-CRE		63782.2	P	 	 	74497.4	P	69139.81	1.17
AFFX-DAP		41.5	A	101.0	A	5.8	A	49.40	0.14
AFFX-LYS		5.9	A	29.0	A	3.8	A	12.90	0.64
AFFX-PHE		8.2	A	7.2	A	26.1	A	13.84	3.19
AFFX-THR		8.9	A	5.6	A	6.3	A	6.95	0.71
AFFX-TRP		55.6	A	73.7	A	31.3	A	53.54	0.56
AFFX-R2-EC-BIOB	3101.7	P	3812.0	P	2961.1	P	3291.59	0.95
AFFX-R2-EC-BIOC	7585.2	P	 	 	8284.0	P	7934.62	1.09
AFFX-R2-EC-BIOD	29350.5	P	 	 	38538.8	P	33944.63	1.31
AFFX-R2-P1-CRE	74415.4	P	 	 	94116.9	P	84266.14	1.26
AFFX-R2-BS-DAP	5.0	A	3.3	A	7.5	A	5.28	1.52
AFFX-R2-BS-LYS	10.4	A	3.2	A	97.2	A	36.90	9.36
AFFX-R2-BS-PHE	8.7	A	10.0	A	82.2	A	33.65	9.44
AFFX-R2-BS-THR	6.9	A	21.0	A	12.6	A	13.51	1.84
______________________________________________________________________

